Preliminary Review for Hands On Training Complete Quiz at the end of this session. Test control solutions the same way a blood sample is tested. The ACCU-CHEK Inform manuale II Blood Glucose Monitoring System is intended for testing outside the body (in vitro diagnostic use) and is intended for multiple patient use in professional healthcare settings. AccuChek Inform II Test Strip & Quality Control Material 4. Accu-Chek Inform II Base Unit serial number ready to help assist our customer care group in troubleshooting. The Accu-Chek® Inform II system is a total solution designed to help you provide the best care possible, delivering accurate results in 5 seconds. Blood specimens may be capillary or drawn from indwelling lines that have never contained glucose.
If the Accu-Chek Inform II meter with RF is used in an area with low signal or interferences, it is recommended to install a connected base unit for redundancy. Seamless network integration. How often should I clean Accu Chek inform II? What is the manual version of Accu-Chek inform II? For more information about the ACCU-CHEK Inform II test strip, refer to the package insert for instructions for use, performance characteristics, and limitations of the system.
ACCU-CHEK Inform II Meter Uses laser light to scan operator ID, test strip and control solution bar codes. Only consider Clinically Validated devices listed by Medaval. Code Information Model Number :Device Identifier:All serial numbers below XXXwhere XXX is the first 3 digits of the serial number). The Accu-Chek Inform II system is the first point-of-care blood glucose device to offer truly wireless technology at the meter level. Database contains 1 Accu-Chek Inform II Competency Manuals (available for free online viewing or downloading in PDF): Manual. . Which of these specimens is an acceptable specimen for the Inform II glucose meter?
As a result, they can treat their patients more efficiently. The multiple-patient use ACCU-CHEK Inform II Blood. What is Informe II ACCU? Accu-Chek Inform II Base Unit Hub Operator’s Manual or the “Technical Note” stored on the Accu-Chek Inform II Base Unit as a PDF file (see page 178 of this manual). 0 ACCU-CHEK ® Inform II System. The ACCU-CHEK® Inform II blood glucose meter with its wireless capability ensures that the right test is being performed on the right patient at the right time. Cleaning the outside Wipe with a slightly moist cloth and mild cleaning agent. View online Training manual for Accu-Chek Inform II Measuring Instruments or simply click Download button to examine the Accu-Chek Inform II guidelines offline on your desktop or laptop computer.
0 Page 1 of 2 Revision Date:Filename and PathX:&92;QSE DOCUMENTS&92;LQSM-Finals&92;Point of Care&92;Accu-Chek Glucose Meter&92;I CBG 0003 F2 Point of Care Quiz - Glucose. This system should only be used with single-use, auto-disabling lancing devices. Note on the use of “base unit” in this manual Note on illustrations in this manual Unless otherwise specified, the term “base unit” refers to both the Accu-Chek Inform II Base Unit and the Accu-Chek Inform II Base Unit Light. Glucose meter is used whenever a glucose test is requested. Patient hematocrit should be between 10-65% Lipemic samples (triglycerides) in excess of 1800 mg/dl may produce elevated results Blood concentrations of galactose >15 mg/dl will cause overestimation of blood glucose accu chek inform ii manuale italiano results. The ACCU-CHEK Inform II Base Unit HUB transfers data from the ACCU-CHEK Inform II Base Unit Light and docked ACCU-CHEK Inform II meter to a data management system as an aid in monitoring the effectiveness of glucose control.
Roche Diagnostics. Accu-Chek Inform II Competency Manuals & User Guides. Database contains 8 Accu-Chek Inform II Manuals (available for free online viewing or downloading in PDF): Quick reference manual, Manual, Operator&39;s manual, Training manual. Manual versions 4, 5 and 6 are for Accu-Chek Inform II meters with serial number versions greater than or equal to UU14000000. Or contact Accu-Chek Customer Care for tech- nical assistance. ACCU-CHEK Inform II strips expire on the date printed on the strip vial label. · Complete lists of error codes for Accu-Chek blood glucose meters and the reasons for each code are in the product owner&39;s booklets and online at Acc-CAN Manuel.
Accu-Chek Inform II meter is using malfunctions or looses service. Get your questions answered by an Accu-Chek Customer Care representative. Accu-chek Inform II meter Single-use and auto-disabling lancing device (Safe-T pro lancets) Alcohol swabs Cotton ball or gauze Accucheck Comfort Curve accu chek inform ii manuale italiano test strips Disposable gloves. The ACCU-CHEK Inform II system is meant for in vitrodiagnostic use by healthcare professionals in clinical settings. · Device Classification Name: system, test, blood glucose, over the counter: 510(k) Number: K121679: Device Name: ACCU-CHEK INFORM II BLOOD GLUCOSE MONITORING SYSTEM.
· ACCU-CHEK Inform II test strips have been developed such that there is NO interference due to Maltose. Accu-Chek ACTIVE User italiano Manual Operation & user’s manual (68 pages) Accu-Chek INFORM II Operator&39;s Manual Operator&39;s manual (196 pages) Accu-Chek Aviva Quick Start Manuals Quick start manuals (2 pages) Accu-Chek Performa Combo Getting Started Manual Getting started manual (74 pages) Accu-Chek Aviva Owner&39;s Booklet Owner&39;s booklet (200 pages. Accu-Chek Inform II: Newborn Care Glucose Meter Protocol Action 1 USE OF GLUCOSE METER Glucose determinations are done by a registered nurse certified in the use of the glucose meter.
Cleaning the Accu-Chek Inform II meter after each patient use: a) prevents transmitting contamination from one patient to another b) is a CMS standard c) is a College of American Pathologists (CAP) standard d) must be performed using hospital approved wipes. AccuChek Inform II Meter 3. Accu-Chek Inform II Base Unit The Accu-Chek Inform II Base Unit with new hard-ware and software (REFhas replaced the legacy version (REFin this manual. Support for online purchases and subscriptions is available from 8:00 AM – 5:00 PM ET, Monday – Friday. Wash hands and don personal protective equipment as required by infection control and isolation policies and procedures. “HI” on the ACCU-CHEK Inform II system display indicates that the blood glucose value is over mg/dL, and the result may be the reading range of the meter.
e) All of the above. What is the "Accu Chek" inform II system? Cleaning and Maintenance The Accu-Chek Inform II should be cleaned regularly, ® as per local policy. ACCU-CHEK Inform II test strip vial. on the vial or bottle labels or in the meter 16. Accu-Chek Inform II Manuals & User Guides User Manuals, Guides and Specifications for your Accu-Chek Inform II Blood Glucose Meter, Measuring Instruments.
Accu-ChekInform II. Accu-Chek accu chek inform ii manuale italiano Inform II Operator&39;s Manual. NOTE about reagents: Reagents are not to be used past their expiration date. docx Retain documentation in employee department file Page 3 of 3 17.
Always switch the meter off before cleaning. DO NOT LET LIQUID DRIP INTO THE METER AND TAKE SPECIAL CARE TO KEEP MOISTURE OUT accu chek inform ii manuale italiano OF THE TEST STRIP GUIDE. The ACCU-CHEK Inform II System Operator’s Manual 17. Opened control solutions are ii good for 3 months from the date first opened, and should be labeled with the open date and expiration date. Page 147: Accu-Chek Inform Ii Operator&39;s Manual And Quick Reference Guide. 2 Accu-Chek Inform II Operator Training. The Accu-Chek Inform II system is designed to deliver data with speed and efficiency, helping you provide optimal care.
The Accu-Chek Inform II System Consists of: Accu-Chek Inform II Glucose Controls: Quality Control solutions should be stored at room temperature (15-30°C) – record temperature daily. The Accu-Chek Inform II Blood Glucose Monitoring System is intended for testing outside the body (in vitro diagnostic use) and is intended for multiple patient use in professional healthcare settings. The appropriate pack of Roche test strips must be stored with the Accu-Chek Inform II for which it has been calibrated. ACCU-CHEK Inform II Control Level 1 (Low) and Control Level 2 (High), the lot and range information will be available in meter. Manual version Revision date Changes. Glucose, Whole Blood, Accu-Chek Inform II Quiz C:&92;Documents and Settings&92;erspamed&92;Local Settings&92;Temporary Internet Files&92;Content.
6 mL The performance of this system has not been evaluated in the critically ill. The redundancy of the con- nected base unit allows immediate transmission of patient results when the meter is docked. The Accu-Chek Inform II Control pack contains two control solutions, one for the hypoglycemic range (control solution 1, gray cap) and one for the hyperglycemic range (control solution 2, white cap). ACCU-CHEK Inform II Controls (high and low levels). With the ACCU-CHEK® Inform II system, hospitals are able to transfer patients’ results to their electronic medical records more quickly. Operator ID & ongoing competency 7. 2 Implementation Date: 10:20:08 PM Folder Name: CW&92;Point of Care&92;Whole Blood Glucose This is a controlled document for CW use only.
Cleaning the meter 6. Download Operator&39;s manual of Accu-Chek Inform II Blood Glucose Meter, Measuring Instruments for Free or View it Online on All-Guides. . Chat and Phone support are available for most queries from 8:00 AM – 8:00 PM ET, Monday – Friday. From the wireless meter, patented test strip technology and easy-to-use lancing devices to seamless IT integration, everything works together so.
The system uses ACCU-CHEK® Inform II test strips. User Manuals, Guides and Specifications for your Accu-Chek Inform II Competency Blood Glucose Meter. A breakthrough in point-of-care blood glucose monitoring For more info, please visit us at: The Accu-Chek ® Inform II solution brings laboratory safety standards to bedside glucose testing and help ensure safety for patients and users Infection control with an integrated product design of the meter, the test strips and the lancets Improved clinical decision making with reports of actionable data per user needs. · The Accu-Chek Guide Blood Glucose Monitoring System is intended to be used by a single person and should not be shared. Caution: Because laser light can damage vision, always avoid staring directly into the laser light or shining the laser light into anyone’s eye while scanning a barcode. An overview of the legacy Accu-Chek Inform II Base Unit and technical information have been moved to Appendix E.
AccuChek Inform II System Components 2. Accu-Chek Inform II - Glucose Meter Quiz ***Print as double-sided document*** Version 2. Roche Accu-Chek inform II Operator Training Video.
Any printed copies are uncontrolled unless specified. See the meter Operator’s Manual for specific use instructions. Outlook&92;JZX2V9RP&92;Glucose Whole Blood Accu-Chek Inform II Quiz (2).
-> Aggiornamenti manuale windows 7
-> Picking manual